FDA Notice of Recall of Drager Medical Inc. Emergency Transport Ventilators
Office of Emergency Medical Services
Department of Public Health
Commonwealth of Massachusetts
FDA Notice: Dräger Medical Inc. Recalls Emergency Transport Ventilators Due to a System Error That May Lead to a Halt in Ventilation Therapy
While Massachusetts is not one of the states from which these ventilators have been recalled, we are notifying Massachusetts ambulance services licensed at the Advanced Life Support or Critical Care level, to make sure you are aware of this issue.
Date: February 10, 2016
The U.S. Food and Drug Administration (FDA) has identified this as a Class I recall, the most serious type of recall. Use of these devices may cause serious injuries or death.
•Dräger Oxylog 2000 plus, Oxylog 3000, and Oxylog 3000 Plus Ventilator
•Catalog numbers: 5705080, 2M86300, 2M86965, 5704813, 5704831
•Distribution Dates: April 1, 2007 to December 12, 2015
•Devices Recalled in the U.S.: 117 distributed in Arkansas, Arizona, California, Connecticut, Florida, Georgia, Iowa, Idaho, Illinois, Indiana, Kansas, Kentucky, Louisiana, Maryland, Maine, Michigan, Minnesota, Missouri, Mississippi, Montana, North Carolina, Nebraska, New Hampshire, New Jersey, New York, Ohio, Oklahoma, Oregon, Pennsylvania, South Carolina, Tennessee, Texas, Utah, Virginia, Vermont, Washington, Wisconsin, West Virginia and Wyoming
The Dräger Oxylog Emergency Transport Ventilators provide constant breathing support for adults and children. These ventilators are used in hospitals or during patient transport.
Reason for Recall
Dräger is recalling the Oxylog Emergency Transport Ventilators because an electrical issue may cause the device to stop working if the control knobs (adjustment potentiometers) are not regularly used.
If the device operator does not intervene, the patient may not receive enough oxygen and could suffer serious adverse health consequences, including injury or death.
Who May be Affected
•Health care providers using Dräger Oxylog 2000 plus, Oxylog 3000, and Oxylog 3000 Plus Ventilators
•All patient groups who may be given breathing support with these ventilators
What to Do
Dräger sent a letter to all customers with affected devices on December 21, 2015, informing them of this issue.
The letter provides the following instructions to release the electrical contact resistance in the control knobs:
•Turn the device off
•Rotate all control knobs at least 10 times to the left and right stop (minimum and maximum value)
Customers with questions about this recall may call Dräger at 1-800-543-5047 (press 1 at the prompt, then 2, then 32349).
Customers with questions regarding the operation and/or servicing of Dräger Oxylog ventilators may contact Dräger Service Technical Support at 1-800-543-5047 (press 4 at the prompt).
Date Recall Initiated: December 22, 2015
How do I report a problem?
Health care professionals and consumers may report adverse reactions or quality problems they experienced using these devices to MedWatch: The FDA Safety Information and Adverse Event Reporting Program either online, by regular mail or by FAX.
Read the FDA notice of this recall, at: http://www.fda.gov/MedicalDevices/Safety/ListofRecalls/ucm485790.htm?source=govdelivery&utm_medium=email&utm_source=govdelivery